|Ondřej Hruška e423303906 add sallen key preset 2||10 months ago|
|gex @ 3a5ac05d9d||10 months ago|
|.gitignore||10 months ago|
|.gitmodules||10 months ago|
|LICENSE||10 months ago|
|README.md||10 months ago|
|UNITS.INI||10 months ago|
|bode1_amplitude.png||10 months ago|
|bode1_phase.png||10 months ago|
|main.py||10 months ago|
GEX demo measuring the amplitude and phase frequency response of analog filtes.
This demo uses an external waveform generador AD9833 (on a breakout module from eBay). It could also use the built-in DAC, but the performance is sub-optimal and the plots obtained with it had a lot of glitches. Feel free to adjust the script to use the internal DAC and try it.
The AD9833 is connected to SPI on pins PA4 (SS), PA5 (SCK), and PA7 (MOSI). Pins PA1 and PA2 are used by the ADC and attach before and after the filter (DUT).
The generator has low output levels (0.65 V max). To better utilize the input range of the ADC, I used OP213 in a non-inverting configuration (resistors 7.5k and 2.2k), with a 10M resistor from the input to GND. This gives us gain of around 4.4, enough to reach just under 3V.
git clone --recursive, or use
git submodule init,
git submodule update
Dependencies - pyplot, numpy, and whatever you need to use the GEX library: typically pyusb or pyserial.